Views:10 Author:Site Editor Publish Time: 2020-08-28 Origin:Site
The Desktop Vacuum Autoclave is the principle of using microorganisms to kill microorganisms. The equipment is designed to remove the cold air in the inner chamber according to the specified sterilization process, and the saturated hot and humid steam is used as the sterilization factor at high temperature and high pressure. In a high-humidity environment, sterilization of items that can be penetrated by steam is achieved according to a combination of pressure and time. The sterilizer consists of a sterilizer body, a sealing door, a pipe system, and a control system. The door opening direction is upward, and the inner chamber material is made of SUS304 stainless steel plate.
Pressure steam sterilizer for medical equipment, hygrometer sterilization and other health materials.
This device can be used normally under the following conditions:
Ambient temperature: 5 ℃ ~ 40 ℃;
Relative humidity: not more than 85%;
Atmospheric pressure: 70 kPa ~ 106 kPa;
Note: The use of sterilizers by manufacturers and users should consider the effect of local atmospheric pressure on sterilizer parameter settings.
Use of power AC: 220V , 50Hz ;
Avoid heavy dust, oil mist, containing conductive particles, corrosive gases, flammable gas environment.
Avoid easily shock or vibration of the occasion.
Avoid high temperature and humidity or easily wet place.
Avoid strong magnetic environment.
The device is equipped with a total of 11 sets of default procedures, including bare equipment, packaging equipment, rubber procedures, dressing procedures, liquid procedures, solid DIY, liquid DIY, medium sterilization are sterile procedures, BD & Helix, The vacuum test belongs to the test program, medium dissolution belongs to the auxiliary class program.
Sterilizer operation procedures include sterilization preparation, sterilization items loading, sterilization operations, sterile items unloading and other steps.