Views:2 Author:Site Editor Publish Time: 2020-12-25 Origin:Site
CENTRIFUGE SAFETYWHEN HAZARDOUS MATERIALS SPILL,WHAT SHOULD WE DO?
Hazardous Material Spil/Exposure
Turn off centrifuge immediately.
Keep centrifuge cover closed for at least 30 minutes
to reduce aerosolization of hazardous material.
Follow spill, exposure, and incident reporting instructions.
For chemicals see here.
For biohazardous materials see the Biosafety ManuaFor radioactive materials:
call the Health Physics Program
How do I dispose of this?
For biohazardous materials.clean/disinfect centrifuge and remove/cross out biohazard sticker. Attach note on centrifuge describing what has been done.For radioactive materials, request radiation safety survey and signage before disposal of centrifuge.
Remember:Conduct rotor or safety cup/bucket cleanup in nearest biosafety cabinet or fumehood, depending on hazard. Use tongs or forceps to avoid contact with sharp-edged debris. Avoid alkaline cleaners for aluminum centrifuge components.Avoid abrasive wire brushes for cleaning.